T&D WebStorage Service is a free cloud-based storage service provided by T&D Corporation. It makes it possible to manage data loggers, monitor current readings and warning status, and download recorded data collectively.
This section explains how to use T&D WebStorage Service with the RTR500B Series Base Unit.
If you already have your User ID, the registration process is not necessary.
The user ID cannot be reissued so please keep it in a safe place. If you lose your user ID, you will not be able to use the data stored in the T&D WebStorage Service.
Access T&D WebStorage Service from a web browser of a computer or mobile device.
Click [Get a User ID] to go to the registration page, and enter the email address and password.
A confirmation email will be sent to the address you entered. Click the link in the email to verify your address and complete the registration process.
Enter the User ID provided in the registration completion notification email to log in to the T&D WebStorage Service.
If you forget your password, you can get a new one reissued by clicking the [Forgotten your Password?] link in the Log In page.
Device registration is required for the RTR500BW. If the Base Unit is an RTR500BC, the registration process is not necessary as it is automatically registered to T&D WebStorage Service after data has been uploaded.
Go to the [Devices] menu and click the [+ Device] button. It will take you to the Add a Device page.
Enter the serial number and registration code for the Base Unit, then click [Add]. If you plan to use more than one RTR500BW Base Unit, you need to repeat this process for each unit.
(If you don't know the registration code, see: [RTR500B Series FAQs]-[General Questions]-[Q. Where can I check the Registration Code for my RTR500BW?] in this Help.)
By registering an RTR500BW, its Remote Units will be automatically registered and will appear when you go back to the [Devices] menu page as shown below.
By uploading data to T&D WebStorage Service from an RTR500BC, the Base Unit and each of the Remote Units will be automatically registered and will appear as shown below.
Click the [Settings] button to view the Remote Unit Info and make Alert Settings.
The Watchdog function is for detecting devices with which communication has not been occurring.
It monitors the communication status of target devices, and when any of them are out of communication for the set time period, a watchdog alert will be issued and an email notification will be sent. To make settings, select the device and set the timeout period to be monitored, and click [Apply].
Current readings data sent from Base Units to T&D WebStorage Service can be viewed from the [Data View] menu.
In the "Registered Devices", you can view data of all registered devices. The "Cluster View" allows you to mix registered devices into viewing groups according to your needs such as by place or purpose.
The Watch List in the [ Dashboard] allows you to check data instantly every time you open the T&D WebStorage Service. You can add up to six devices to the list and view the most recent measurements as well as click the device name to open the graph window.
∗ The current readings sent from RTR500BW includes the most recently recorded data. More detailed information can be found in the following section in this Help.
(See: [RTR500BW for Windows]-[Details of Operations]-[About Sending Current Readings])
You can reset the graph data for each Remote Unit.
Open the [Devices] menu and click the [Settings] button for the target Remote Unit.
Click the [Reset Data] button at the top right of the Remote Unit Info.
The devices in alert status, alert log and watchdog alerts can be viewed from the [Alert View] menu.
About the watchdog function, please refer to Watchdog Settings.
For warning information received from the RTR500BW, you can select categories to be sent in warning emails.
Go to [Devices] menu - RTR500B series, and click [Settings] of the RTR500BW to open the settings information. In the [Warning Mail Transmission Settings], select ON or OFF for each category.
The warning email recipient addresses can be added up to three (except for the email address registered to your user account).
To add the recipient address, go to the [Account] menu - [Set Warning Recipients] to open the setting window.
Enter the E-mail Address and click the [Add] button.
You can change or remove the address in the [Registered Warning Mail Recipient List].
Warning notifications via email from T&D WebStorage Service include the following:
Upper/Lower Limit Exceeded
Battery Level of Remote Unit
Sensor Error (Problem with the sensor such as a malfunction or poor connection)
Warning and recovery occurred within the warning monitoring time
Wireless Communication Error
Recovery from Warning Status
"Time Stamp" refers to the time when a warning condition was detected and the judgment countdown started.
From the [Download] menu, you can download recorded data as well as delete from T&D WebStorage Service.
Further data analysis is possible by using the T&D Graph software in conjunction with T&D WebStorage Service.
With T&D Graph, it is possible to directly access and open the recorded data stored in the T&D WebStorage Service.
T&D Website > Software > T&D Graph Download
Open T&D Graph and go to the [File] Menu - [Web Storage Service].
Enter the user ID and password registered with T&D WebStorage Service, and click [Login].
All data stored in your WebStorage account will be displayed in a list. Right click on the selected recorded data and click [Download] to download for analysis.